- Open Access
Cone-beam computed tomography in hypofractionated stereotactic radiotherapy for brain metastases
© Ingrosso et al; licensee BioMed Central Ltd. 2012
- Received: 14 November 2011
- Accepted: 1 April 2012
- Published: 1 April 2012
To assess interfraction translational and rotational setup errors, in patients treated with image-guded hypofractionated stereotactic radiotherapy, immobilized by a thermoplastic mask and a bite-block and positioned using stereotactic coordinates.
37 patients with 47 brain metastases were treated with hypofractionated stererotactic radiotherapy. All patients were immobilized with a combination of a thermoplastic mask and a bite-block fixed to a stereotactic frame support. Daily cone-beam CT scans were acquired for every patient before the treatment session and were matched online with planning CT images, for 3D image registration. The mean value and standard deviation of all translational (X, Y, Z) and rotational errors (θx, θy, θz) were calculated for the matching results of bone matching algorithm.
A total of 194 CBCT scans were analyzed. Mean +/- standard deviation of translational errors (X, Y, Z) were respectively 0.5 +/- 1.6 mm (range -5.7 and 5.9 mm) in X; 0.4 +/- 2.7 mm (range -8.2 and 12.1 mm) in Y; 0.4 +/- 1.9 mm (range -7.0 and 14 mm) in Z; median and 90th percentile were respectively within 0.5 mm and 2.4 mm in X, 0.3 mm and 3.2 mm in Y, 0.3 mm and 2.2 mm in Z. Mean +/- standard deviation of rotational errors (θx, θy, θz) were respectively 0.0 degrees+/- 1.3 degrees (θx) (range -6.0 degrees and 3.1 degrees); -0.1 degrees +/- 1.1 degrees (θy) (range -3.0 degrees and 2.4 degrees); -0.6 degrees +/- 1.4 degrees (θz) (range -5.0 degrees and 3.3 degrees). Median and 90th percentile of rotational errors were respectively within 0.1 degrees and 1.4 degrees (θx), 0.0 degrees and 1.2 degrees (θy), 0.0 degrees and 0.9 degrees (θz). Mean +/- SD of 3D vector was 3.1 +/- 2.1 mm (range 0.3 and 14.9 mm); median and 90th percentile of 3D vector was within 2.7 mm and 5.1 mm.
Hypofractionated stereotactic radiotherapy have the significant limitation of uncertainty in interfraction repeatability of the patient setup; image-guided radiotherapy using cone-beam computed tomography improves the accuracy of the treatment delivery reducing set-up uncertainty, giving the possibility of 3-dimensional anatomic informations in the treatment position.
- Brain Metastasis
- Plan Target Volume
- Gross Tumour Volume
- Stereotactic Radiotherapy
- Rotational Error
Brain is a common site for metastases. Stereotactic radiosurgery (SRS) and hypofractionated stereotactic radiotherapy (HSRT), due to their ability to deliver very high doses to a small volume with high tumour control rates , may be considered as a standard treatment for patients with brain metastases. From a biological point of view HSRT might have some advantage in comparison to SRS in terms of acute complications  and of tumour control rate for lesions larger than 10 cc (or more than 3 cm of diameter) [3, 4]. In HSRT, daily treatment reproducibility is necessary in order to avoid geographic miss of the target; as brain metastases are not affected by internal organ motion and their position can be considered stably correlated with bony structures, patient set-up is the crucial step for the exact treatment delivery. Several noninvasive systems are used for patient immobilization, but verification of the patient position is always necessary to enhance the precision of the stereotactic treatment. Image-guided radiotherapy (IGRT) using cone-beam computed tomography (CBCT) offers the possibility of a daily precise detection and correction of translational and rotational set-up errors.
The aim of this study was to assess inter-fraction translational and rotational set-up errors, in patients treated with IGRT-HSRT immobilized by a thermoplastic mask and a bite-block and positioned using stereotactic coordinates.
Patient and treatment characteristics
Planning target volume
total dose and
18 Gy (3 × 6 Gy)
24 Gy (4 × 6 Gy)
30 Gy (2 × 15 Gy)
30 Gy (5 × 6 Gy)
32 Gy (4 × 8 Gy)
40 Gy (5 × 8 Gy)
where n is the number of fractions, d is the fraction dose.
Patients were treated with Linac Elekta Synergy® S (Elekta Oncology Systems, Crawley, UK) equipped with Beam ModulatorTM (leaf width of 0.4 cm at the isocenter) and a kilovolt (kV) imaging system capable of acquiring 3D X-ray volume images based on kV cone-beam computed tomography (CBCT). In particular CBCT has the tube and flat panel imager both mounted on retractable arms that extend from the accelerator's drum structure. The kilovoltage system is mounted in an orthogonal direction to the MV system sharing a common axis of rotation. We used the "head and neck protocol" with the following parameters: 100 kV, 36.1 mAs, nominal scan dose 0.9 mGy, FOV (Field Of View) at the isocenter 260.4 mm. All the images were stored and processed on a control work station (XVI).
For every patient, planning CT images, with OARs, PTV and marker isocenter were transferred from Pinnacle to XVI.
Daily CBCT scans were acquired for every patient before the treatment session. Projections are acquired during a single rotation of the gantry (about 113 s) and processed with Elekta XVI software (XVI, Elekta, Crawley, United Kingdom) that with a back-projection algorithm reconstructs the 3D volumetric images. Reconstruction of about 625 projections is performed on a Intel Xeon (TM) 3.06 GHz processor and it takes about 30 s to reconstruct a 410 × 264 × 410 voxel matrix with voxel dimension 1 × 1 × 1 mm3.
Planning CT images were matched online with the daily CBCT images using bone-matching algorithm for 3D image registration (chamfer matching).
An alignment clip-box for volumes matching was defined by the physicians. The registration was checked by a physician using a "cut" display modality. After 3D registration the XVI software calculates translational and rotational set-up errors. On-line corrections were performed before each treatment. The translational errors were corrected using the mechanical movements of the couch; the rotational errors were corrected using the knobs of the stereotactic head frame support.
If the translational and rotational errors in the first set-up exceeded respectively 6 mm and/or 3 degrees, the patient was repositioned.
The mean value and standard deviation of all translational (X, Y, Z) and rotational errors (θx, θy, θz) were calculated for the matching results of bone matching algorithm. It was calculated the length of the translational correction 3D vector using the formula3D vector = (X2+Y2+Z2)1/2.
To test the reproducibility of isocenter detection and position correction an intense program of quality assurance tests was performed: a home made phantom and an anthropomorphic phantom were used to assess error correction performance .
For the 194 CBCT, mean, median, standard deviation (SD), 90th percentile and range of set-up errors were calculated along the three major directions [lateral (X), cranio-caudal (Y) and antero-posterior (Z)] considering the totality of the data obtained for all patients and all fractions. The same parameters were evaluated for rotational set-up errors (θx), (θy), (θz).
Patient repositioning errors (194 CBCT)
Median and 90th percentile of translational errors were respectively within 0.5 mm and 2.4 mm (X axis), 0.3 mm and 3.2 mm (Y axis), 0.3 mm and 2.2 mm (Z axis).
Mean ± SD of rotational errors (θx, θy, θz) were respectively 0.0° ± 1.3° (θx) (range -6.0° and 3.1° mm); -0.1° ± 1.1° (θy) (range -3.0° and 2.4°); -0.6° ± 1.4° (θz) (range -5.0° and 3.3°).
Median and 90th percentile of rotational errors were respectively within 0.1° and 1.4° (θx), 0.0° and 1.2° (θy), 0.0° and 0.9° (θz). Mean ± SD of 3D vector was 3.1 ± 2.1 mm (range 0.3 and 14.9 mm); median and 90th percentile of 3D vector was within 2.7 mm and 5.1 mm.
Stereotactic radiotherapy is a treatment modality for brain metastases to deliver very high doses to a small volume and to obtain high tumour control rates. Hypofractionated stereotactic radiotherapy has the advantage to deliver large doses - probably more effective at killing radioresistant tumours- with the radiobiological features of fractionation [6, 7]; in fact, from a biological point of view it seems to be better than SRS in terms of acute complications  and of tumour control rate for lesions larger than 10 cc [3, 4]. To avoid geographic miss of the target and to spare organs at risk from irradiation, high precision in the daily set-up is required. In stereotactic radiosurgery this precision can be achieved using an invasive rigid fixation of the skull to a stereotactic system that is placed for immobilization before the planning, and it is removed at the end of the treatment delivery. In fractionated stereotactic radiotherapy we use non invasive fixation, such as bite-block and/or a mask system, introducing a significant variation in daily set-up. The uncertainties, related to the non invasive fixation system, can result from different causes such as weight changes of the patient (weight gain for patients receiving steroids), different mask-making technique, thermoplastic mask shrinking .
The combination of two relocatable immobilization devices, such as thermoplastic mask and bite-block, is more effective for head treatment-position reproducibility. Recently, Masi et al. evaluated set-up errors measured by a CBCT for patients immobilized by a thermoplastic mask and a bite-block; when patients were completely repositioned, the set-up corrections were significantly larger than those measured when patients were left immobilized, and set-up errors measured with mask alone were larger than those obtained with a mask and a bite-block, but not with statistically significant difference. In the analysis of 131 CBCT for 57 patients receiving SRT, they obtained an overall mean and standard deviation for the 3D vector of 3.0 ± 1.4 mm and a maximum standard deviation of 2.4 mm that was observed along the cranio-caudal direction .
In the study of Boda-Heggemann et al.  the accuracy of two different mask systems (rigid mask vs thermoplastic mask), using CBCT three-dimensional matching, was compared; the mean module of 3D displacement vector was 3.1 mm ± 1.5 mm for patients immobilized with rigid mask and 4.7 ± 1.7 mm for those immobilized with the thermoplastic mask; Guckenberger et al. measured a mean three-dimensional set-up error of 4.0 ± 2.1 mm according to the bony anatomy of the skull, in 18 patients treated with image-guided stereotactic radiotherapy using kv-CBCT. All these patients underwent a CBCT and a conventional CT after injection of i.v. contrast, just before the treatment session. Set-up errors using automatic bone registration (CBCT) and manual soft tissue registration of brain metastases (conventional CT) were compared, demonstrating a significant correlation between the two types of registration (r ≥ 0.88) and thus the great accuracy of daily repositioning with image-guidance based on the bony anatomy of the skull that can be used as a surrogate for the actual target position .
In the present study we evaluated set-up accuracy using automatic image registration between on-board kV CBCT and planning CT scan, based on a clipbox involving the base of the skull; a system based on mask and bite-block immobilization was used. A total of 194 CBCT have been examined. We obtained an overall mean and standard deviation for the 3D vector of 3.1 ± 2.1 mm, that is comparable with data obtained by other authors, with a range of the 3D vector between 0.3 mm and 14.9 mm; we registered the widest range of set-up errors in the cranio-caudal direction (between -7 mm and 14 mm). The analysis of daily repositioning errors shows that errors related to mask-making technique and patient compliance have to be considered for a stereotactic treatment; in our series in 17 cases we had a translational error exceeding 6 mm and/or a rotational error > 3 degrees and we repeated all the set-up procedure assuming that this difference was attributable to the wrong position of the head into the mask, most of all due to the movements of the joint between the occipital bone and the first cervical vertebra, or to radiation therapist random errors.
The accuracy of automatic 3D-3D matching for evaluation of patient set-up is better than manual 2D-2D matching; for Elekta XVI automatic registration, phantom studies have shown errors less than 0.5 mm [12, 13]. The fractionated stereotactic radiotherapy needs IGRT on-line corrections; on-board Kv CBCT, acquiring volumetric images inside the treatment room, allows daily detection and correction of systematic and random errors before each treatment session, with a small-dose exposure and good image quality [14–17].
In our analysis stereotactic fractionated radiotherapy have the significant limitation of uncertainty in inter-fraction repeatability of the patient set-up; IGRT using cone-beam computed tomography improves the accuracy of the treatment delivery reducing set-up uncertainty, giving the possibility of 3-dimensional anatomic informations in the treatment position.
- Larson DA, Bova F, Eisert D, et al.: Current radiosurgery practice: Results of an ASTRO survey. Int J Radiat Oncol Biol Phys 1994, 28: 523-526. 10.1016/0360-3016(94)90080-9View ArticlePubMedGoogle Scholar
- Hall EJ, Brenner DJ: The radiobiology of radiosurgery: Rationale for different treatment regimens for AVMs and malignancies. Int J Radiat Oncol Biol Phys 1993, 25: 381-385. 10.1016/0360-3016(93)90367-5View ArticlePubMedGoogle Scholar
- Shiau CY, Sneed KP, Shu HKG, et al.: Radiosurgery for brain metastases: Relationships of dose and pattern of enhancement to local control. Int J Radiat Oncol Biol Phys 1997, 37: 375-383. 10.1016/S0360-3016(96)00497-XView ArticlePubMedGoogle Scholar
- Chen JCT, Petrovich Z, O'Day S, et al.: Stereotactic radiosurgery in the treatment of metastatic disease to the brain. Neurosurgery 2000, 47: 268-281. 10.1097/00006123-200008000-00003View ArticlePubMedGoogle Scholar
- Falco MD, Fontanarosa D, Miceli R, Carosi A, Santoni R, D'Andrea M: Preliminary studies for a CBCT imaging protocol for offline organ motion analysis: registration software validation and CTDI measurements. Med Dosim 2011,36(1):91-101. Epub 2010 Apr 3 10.1016/j.meddos.2010.01.003View ArticlePubMedGoogle Scholar
- Brenner DJ, Martel MK, Hall EJ: Fractionated regimens for stereotactic radiotherapy of recurrent tumors in the brain. Int J Radiat Oncol Biol Phys 1991, 21: 819-824. 10.1016/0360-3016(91)90703-7View ArticlePubMedGoogle Scholar
- Giubilei C, Ingrosso G, D'Andrea M, Benassi M, Santoni R: Hypofractionated stereotactic radiotherapy in combination with whole brain radiotherapy for brain metastases. J Neurooncol 2009,91(2):207-12. 10.1007/s11060-008-9700-8View ArticlePubMedGoogle Scholar
- Tsai JS, Engler MJ, Ling MN, et al.: A non-invasive immobilization system and related quality assurance for dynamic intensity modulated radiation therapy of intracranial and head and neck disease. Int J Radiat Oncol Biol Phys 1999, 43: 455-467. 10.1016/S0360-3016(98)00398-8View ArticlePubMedGoogle Scholar
- Masi L, Casamassima F, Polli C, Menichelli C, Bonucci I, Cavedon C: Cone beam CT image giudance for intracranial stereotactic treatments: comparison with a frame guided set-up. Int J Radiat Oncol Biol Phys 2008, 71: 926-933. 10.1016/j.ijrobp.2008.03.006View ArticlePubMedGoogle Scholar
- Boda-Heggemann J, Walter C, Rahn A, et al.: Repositioning accuracy of two different mask systems-3D revisited: comparison using true 3D/3D matching with cone-beam CT. Int J Radiat Oncol Biol Phys 2006, 66: 1568-1575. 10.1016/j.ijrobp.2006.08.054View ArticlePubMedGoogle Scholar
- Guckenberger M, Baier K, Guenther I, et al.: Reliability of the bony anatomy in image-guided stereotactic radiotherapy of brain metastases. Int J Radiat Oncol Biol Phys 2007, 69: 294-301. 10.1016/j.ijrobp.2007.05.030View ArticlePubMedGoogle Scholar
- Oldham M, Letourneau D, Watt L, et al.: Cone-beam-CT guided radiation therapy: A model for on-line application. Radiother Oncol 2005, 75: 271-278.View ArticlePubMedGoogle Scholar
- Meyer J, Wilbert J, Baier K, et al.: Positioning accuracy of cone-beam computed tomography in combination with a HexaPOD robot treatment table. Int J Radiat Oncol Biol Phys 2007, 67: 1220-1228. 10.1016/j.ijrobp.2006.11.010View ArticlePubMedGoogle Scholar
- McBain CA, Henry AM, Sykes J, et al.: X-ray volumetric imaging in image-guided radiotherapy the new standard in ontreatment imaging. Int J Radiat Oncol Biol Phys 2006, 64: 625-634. 10.1016/j.ijrobp.2005.09.018View ArticlePubMedGoogle Scholar
- Sykes J, Amer A, Czajka J, et al.: A feasibility study for image guided radiotherapy using low dose, high speed, cone beam X-ray volumetric imaging. Radiother Oncol 2005, 77: 45-52. 10.1016/j.radonc.2005.05.005View ArticlePubMedGoogle Scholar
- Letourneau D, Wong JW, Oldham M, et al.: Cone-Beam-CT guided radiation therapy: Technical implementation. Radiother Oncol 2005, 75: 279-286. 10.1016/j.radonc.2005.03.001View ArticlePubMedGoogle Scholar
- Islam MK, Purdie TG, Norrlinger BD, et al.: Patient dose from kilovoltage cone beam computed tomography imaging in radiation therapy. Med Phys 2006, 33: 1573-1582. 10.1118/1.2198169View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.