Evaluation of heterogeneity dose distributions for Stereotactic Radiotherapy (SRT): comparison of commercially available Monte Carlo dose calculation with other algorithms
© Takahashi et al; licensee BioMed Central Ltd. 2012
Received: 12 September 2011
Accepted: 9 February 2012
Published: 9 February 2012
The purpose of this study was to compare dose distributions from three different algorithms with the x-ray Voxel Monte Carlo (XVMC) calculations, in actual computed tomography (CT) scans for use in stereotactic radiotherapy (SRT) of small lung cancers.
Slow CT scan of 20 patients was performed and the internal target volume (ITV) was delineated on Pinnacle3. All plans were first calculated with a scatter homogeneous mode (SHM) which is compatible with Clarkson algorithm using Pinnacle3 treatment planning system (TPS). The planned dose was 48 Gy in 4 fractions. In a second step, the CT images, structures and beam data were exported to other treatment planning systems (TPSs). Collapsed cone convolution (CCC) from Pinnacle3, superposition (SP) from XiO, and XVMC from Monaco were used for recalculating. The dose distributions and the Dose Volume Histograms (DVHs) were compared with each other.
The phantom test revealed that all algorithms could reproduce the measured data within 1% except for the SHM with inhomogeneous phantom. For the patient study, the SHM greatly overestimated the isocenter (IC) doses and the minimal dose received by 95% of the PTV (PTV95) compared to XVMC. The differences in mean doses were 2.96 Gy (6.17%) for IC and 5.02 Gy (11.18%) for PTV95. The DVH's and dose distributions with CCC and SP were in agreement with those obtained by XVMC. The average differences in IC doses between CCC and XVMC, and SP and XVMC were -1.14% (p = 0.17), and -2.67% (p = 0.0036), respectively.
Our work clearly confirms that the actual practice of relying solely on a Clarkson algorithm may be inappropriate for SRT planning. Meanwhile, CCC and SP were close to XVMC simulations and actual dose distributions obtained in lung SRT.
Lung cancer is the leading cause of cancer-related death for men in the world , and there are no indications of a decrease in the number of these mortalities. Recently, because of the application of computed tomography (CT) for lung cancer screening, the number of patients diagnosed with early-stage non-small cell lung cancer (NSCLC) has increased. At present, stereotactic radiation therapy (SRT) is considered as a therapeutic option in early stage lung cancer for inoperable patients or for patients refusing surgery. SRT has the advantage of being a minimally-invasive procedure, as well as having a relatively short duration of the course of treatment. In addition, SRT continues to show much better outcomes than those of conventional radiotherapy [2, 3].
Since lung is the most inhomogeneous site in the human body, it is very important for SRT planning to take into account differences in tissue density in the dose computation and to consider the secondary electron transport accurately. Therefore, the use of heterogeneity correction and different types of algorithms have been reported to significantly influence the accuracy of the absolute dose [4, 5].
To confirm safety and efficacy, SRT for lung cancer is still under evaluation in multi-institutional clinical trials. For example, the Japan Clinical Oncology Group (JCOG) conducted a phase II study 0403 of SRT in operable and medically inoperable patients with pathologically proven T1N0M0 non-small cell lung cancer (NSCLC) to evaluate efficacy and safety. Patient accrual for operable cases and their 3-year follow up was completed in February 2010 . Moreover, JCOG 0702, a phase I dose escalation study of SRT in patients medically inoperable or unfit for surgery with pathologically proven T2N0M0 NSCLC, was started to determine the recommended dose. In this context, in JCOG 0403, the prescribed dose was 48 Gy at the isocenter in 4 fractions and heterogeneity corrected doses by pencil beam convolution (PBC) algorithms were used since PBC could commonly be used in almost all clinical practices at that time. However, at the present time it is well known that PBC has shortcomings when it comes to severe inhomogeneities [7, 8]. As for lung cancer treatments, the actual dose is to be lower than expected. In JCOG 0702, therefore, the prescription was changed and the planning objective was for 95% of the PTV to be covered by the some isodose (i.e., 50 Gy) with superposition (SP) or other newer algorithm than Clarkson.
Several studies have been conducted on the accuracy of inhomogeneity corrections employing various algorithms. Most of these studies, however, focus on evaluating the dose to the phantom [9, 10]. Koelbel et al.  found that the Pencil Beam (PB) algorithm overestimates the dose to targets in the lung as compared to CCC. These reports were not examined in any real patient. Fragoso et al.  reported that XVMC is more accurate than PB in SRT. This report, however, compared only two algorithms and was limited for only three patients.
In contrast, the present work summarized four different dose calculations of actual SRT cases with 20 lung cancer patients after dosimetry measurements on an inhomogeneous phantom. The following four dose-calculation algorithms were compared: scatter homogeneous mode (SHM) and CCC available in Pinnacle3 (Philips©), SP implemented in XiO (ELEKTA©), and XVMC implemented in Monaco (ELEKTA©). This study did not include both PB and Clarkson algorithms. Instead, the SHM calculation corresponding Clarkson calculation was employed. Sharing a common plan, dose volume histograms (DVHs) calculated by these algorithms were analyzed to quantify the dose to the targets and lung as an organ at risk (OAR). Furthermore, mean dose, and mean relative difference were employed to assess the algorithms.
Characteristics of 20 patients
CT images for treatment planning were acquired using Aquillion™LB (TOSHIBA©) after patients were positioned in a stereotactic body frame (SBF; ELEKTA©) in the supine position. An SBF-attached template on the patient's abdomen reduced the mobility of the target . The CT images were acquired with one mm thick slices around the tumor and five-mm-slices elsewhere using the "long-scan-time" technique, which visualized a major part of the trajectory of tumor movement by scanning each slice for a long time . Slow CT scan was performed during four seconds with abdominal compression.
The ITV was delineated on a three-dimensional radiation treatment planning system (3D RTPS) (Pinnacle3 Version 7.4i, Philips©) using the lung window of the CT scan for treatment planning. For each case, ITVs and OARs were defined manually slice by slice by the same radiation oncologist. In all cases, planning target volumes (PTVs) were created by adding five mm margins to the ITVs in all directions, and generated using the automatic three-dimensional (3D) contour generation tool of Pinnacle3. All calculations were performed with a grid size of 2.0 mm.
Since the JCOG 0403 protocol has involved the older algorithms, such as the PB and Clarkson algorithm, all plans were first calculated with a SHM calculation ignoring a heterogeneous scatter in Pinnacle3. The planned dose was 48 Gy in four fractions, using static beams (eight ports); the gantry and couch angles were 180° + 0°, 260° + 0°, 340° + 0°, 30° + 40°, 35° + 320°, 320° + 320°, 30° + 90°, 330° + 90°, respectively. All plans had 6 MV non-coplanar and non-opposing rectangular beams.
In the second step, the CT images and structures, such as PTV and organs at risk (OAR), and beam data were exported to the Xio TPS and the Monaco TPS. Special care was taken to preserve the same exact plan (keeping the monitor units (MUs), beam weights and fixed angles) in all algorithms. Then, three other dose algorithms were used for recalculating: collapsed cone convolution (CCC) from Pinnacle3 ver.7.4i, and superposition from XiO ver.4.4 (ELEKTA©), and XVMC from Monaco ver.2.03.01 (ELEKTA©).
In Pinnacle CCC dose calculation, to account for inhomogeneity, the kernels are density-scaled during superposition that is performed by using "collapsed cones", which refer to the modeling of a cone in space using a single ray corresponding to the central axis of the cone. The polyenergetic kernels were constructed with Monte Carlo-generated energy deposition cones [15, 16]. In the Xio superposition dose calculation method, a fanned grid is created and dose is computed by convolving the total energy released with Monte Carlo-generated energy deposition kernels which are represented in spherical coordinates . For more accurate calculation in inhomogeneous tissues the kernels are allowed to change with the local electron density variations and to reduce calculation time by using the multi-grid superposition method . The XVMC dose engine implemented in Monaco (ELEKTA©) was applied in this study. Kawrakow et al.  developed VMC as a fast calculation engine for electron beams, and VMC was later extended to photon beams as XVMC . The dose calculation cube voxel size and the statistical uncertainty of the XVMC dose calculation in this study were 2 × 2 × 2 mm3 and 1%, respectively. All of the TPS's were installed in the Department of Radiology, University of Tokyo hospital, and were commissioned for a 6 MV photon beam provided by an Synergy linear accelerator (ELEKTA©).
To validate the reliability of the comparison among various algorithms, it is required to verify the accuracy of beam modeling in each TPS. After a commissioning recommended by vendors, therefore, the verification of each TPS for a small field size was performed on a water phantom (RT-3000-New-Water, R-tech, Japan) and an inhomogeneous phantom with a spherical insert of 3 cm diameter (RT-3000-New-Water with cork, R-tech, Japan), which was placed on the middle of the cork. The isocenter was located on the middle of these phantoms. All calculations were performed with volume for a grid size 2 × 2 × 2 mm3 by using the actual CT images with a CT-to-density table of Aquillion™LB. The comparison was made with a 3 × 3 cm2 single beam in the water phantom and the same eight non-coplanar beams of 4 × 4 cm2 as shown in previous subsection in the inhomogeneous phantom. These results were compared with the measurement at isocenter using a pinpoint ion chamber (PTW, Germany).
Treatment plan evaluation
The dose distributions were compared with each other. This study also evaluated the shape differences of the dose volume histograms (DVHs) for the different calculations using the same plan. For each treatment plan, the minimal, maximal, and median relative doses in the ITV and PTV were compared and the differences calculated. The paired t- test was used to examine the differences in dose volume indices between different algorithms. A probability value of less than 0.05 was considered to be significant.
Comparison of IC doses in four algorithms with the measeured doses
Single Beam (3 × 3 cm2) in Water Phantoma)
Eight Beams (4 × 4 cm2) in inhomogeneous Phantomb)
Evaluation of dose distributions in 20 patients
Characteristics and dose-volume parameters of the different dose calculations
CCC vs. SP
CCC vs. MC
SP vs. MC
Comparison of dose-volume histograms of SHM, CCC, SP and XVMC
The dose distributions for the ITV, PTV, and OAR structures show differences between the four algorithms, and this can be appreciated in the DVH in case 13. Figure 3 shows the DVH comparison of calculations with SHM, CCC, SP and XVMC. The blue lines are for lung dose (Lung-PTV), the black lines for ITV, and the red lines for PTV.
As can be seen in Figure 3, SHM had a tendency to produce higher dose distributions in ITV and PTV than the other algorithms. On the other hand, lung doses (bilateral lung volume minus ITV) were almost the same among all four algorithms. While SHM showed significant differences compared to XVMC in the doses of ITV and PTV, CCC and SP both had similar dose volume histograms to XVMC.
Differences in dose volume indices
Table 3 lists the mean values and standard deviations of the relative differences in several dose volume indices between the different algorithms. The dose discrepancies between SHM and the others for the median values of the IC dose were -5.5% (median, range: -4.1 to -6.2%) and of the PTV95 were -9.8% (range: -7.8 to -11.7%). Compared to XVMC, SHM greatly overestimated the doses in IC doses and PTV95. The differences in mean doses were 3.16 Gy (6.6%) for IC doses, and 5.24 Gy (10.9%) for PTV95. From this study, an isocentric clinical dose of 48 Gy with Clarkson-type calculation in JCOG 0403 was equivalent to IC 45.4 Gy with CCC, 46.0 Gy with SP and 44.84 Gy with XVMC.
Differences were noted in IC doses between SP and XVMC (-2.67%, p = 0.0036), and in PTV95 between SP and XVMC (-4.39%, p = 0.033). We also observed differences in ITV min doses between SP and XVMC (-4.2%, p = 0.0094). In contrast, we observed no significant differences between CCC and XVMC, and between CCC and SP. No significant differences were found among the four algorithms in conformity index (CI, which is the treated volume, defined as the volume enclosed by the isodose curve of the PTVmin, divided by the PTV volume), homogeneity index (HI, which is the maximal dose divided by the minimal dose), mean lung dose (MLD), V20 (percentage of volume covered by the 20 Gy isodose line) and PTV min.
For SRT planning, accuracy in the dose calculation is required because of the heterogeneity in lung tissue. It is acknowledged that CCC can accurately predict the dose distribution . In addition, XVMC methods are potentially the most powerful dose calculation algorithms in radiotherapy . However, due to long calculation times and the huge computational power needed, XVMC algorithms have not seen widespread clinical use . On our TPS, the CCC algorithm (Pinnacle3 ver.7.4i) is approximately 2.5 times faster than XVMC (Monaco ver.2.03) with a small loss in accuracy (10 min vs. 25 min). In addition, we found overall that in SRT planning, the CCC and SP algorithm can reproduce the dose distribution calculated by XVMC very well. Thus, they remain attractive options for routine use in the hospital due to their short computation times.
It is well known that target dose tends to be lower with CCC than with Clarkson. Generally, this implies that pencil beam-like algorithms such as SHM tend to give the wrong impression that a good PTV coverage has been achieved when in reality this is not the case. The reason for this is lateral electron scattering, which is neglected by Clarkson . Therefore, simple algorithms such as SHM especially overestimate the dose in the interface between the target and lung tissue . Our findings have shown agreement with a previous study [9, 25]. From this study, an isocentric clinical dose of 48 Gy with the Clarkson algorithm in JCOG 0403 was equivalent to PTV95 dose of 40.5 Gy with CCC in JCOG 0702 (data not shown).
It is concluded that CCC and SP delivered almost the same dose distribution when used in combination open fields like SRT for lung cancer. Our analysis and interpretation is deficient in needing further investigation to ascertain whether CCC and SP are preferable to manage leakage and radiation quality in IMRT.
This is the first study comparing XVMC with SHM, CCC, and SP for lung SRT treatment plans. The dose distributions in actual CT scans from CCC and SP almost agreed with those of XVMC at energy of 6 MV. An IC dose of 48 Gy with the Clarkson-type algorithm in JCOG 0403 was equivalent to PTV95 dose of 40.5 Gy with CCC in JCOG 0702. We should take careful note of the interpretational problems arising from this discrepancy. As reported in previous studies, our work clearly confirms that the actual practice of relying solely on Clarkson-type algorithm may be inappropriate for SRT planning. The CCC and SP were close to XVMC simulations which we assumed to be the best representation of the real dose distributions in lung SRT. Therefore, CCC and SP are still reliable methods for SRT of lung tumors.
- Jemal A, Bray F, Center MM, Ferlay J, Ward E, Forman D: Global cancer statistics. CA Cancer J Clin 2011,61(2):69-90. Epub 2011 Feb 4 10.3322/caac.20107View ArticlePubMedGoogle Scholar
- Matsuo Y, Shibuya K, Nagata Y, Takayama K, Norihisa Y, Mizowaki T, Narabayashi M, Sakanaka K, Hiraoka M: Prognostic Factors in Stereotactic Body Radiotherapy for Non-Small-Cell Lung Cancer. Int J Radiat Oncol Biol Phys 2010. Epub 2010 May 14Google Scholar
- Baumann P, Nyman J, Hoyer M, Wennberg B, Gagliardi G, Lax I, Drugge N, Ekberg L, Friesland S, Johansson KA, Lund JA, Morhed E, Nilsson K, Levin N, Paludan M, Sederholm C, Traberg A, Wittgren L, Lewensohn R: Outcome in a prospective phase II trial of medically inoperable stage I non-small-cell lung cancer patients treated with stereotactic body radiotherapy. J Clin Oncol 2009,27(20):3290-6. Epub 2009 May 4 10.1200/JCO.2008.21.5681View ArticlePubMedGoogle Scholar
- Herman Tde L, Gabrish H, Herman TS, Vlachaki MT, Ahmad S: Impact of tissue heterogeneity corrections in stereotactic body radiation therapy treatment plans for lung cancer. J Med Phys 2010,35(3):170-3. 10.4103/0971-6203.62133View ArticlePubMedGoogle Scholar
- Chang D, Liu C, Dempsey JF, Palta JR, Kopea J, Louis D, Morris C, Chopra R, Olivier KR: Predicting changes in dose distribution to tumor and normal tissue when correcting for heterogeneity in radiotherapy for lung cancer. Am J Clin Oncol 2007,30(1):57-62. 10.1097/01.coc.0000251222.36417.3bView ArticlePubMedGoogle Scholar
- Hiraoka M, Ishikura S: A Japan clinical oncology group trial for stereotactic body radiation therapy of non-small cell lung cancer. J Thorac Oncol 2007,2(7 Suppl 3):S115-7.View ArticlePubMedGoogle Scholar
- Knoos T, Ahnesjo A, Nilsson P, Weber L: Limitations of a pencil beam approach to photon dose calculations in lung tissue. Phys Med Biol 1995, 40: 1411-1420. 10.1088/0031-9155/40/9/002View ArticlePubMedGoogle Scholar
- Aarup LR, Nahum AE, Zacharatou C, Juhler-Nøttrup T, Knöös T, Nyström H, Specht L, Wieslander E, Korreman SS: The effect of different lung densities on the accuracy of various radiotherapy dose calculationmethods: implications for tumour coverage. Radiother Oncol 2009,91(3):405-14. Epub 2009, 16 10.1016/j.radonc.2009.01.008View ArticlePubMedGoogle Scholar
- Dobler B, Walter C, Knopf A, Fabri D, Loeschel R, Polednik M, Schneider F, Wenz F, Lohr F: Optimization of extracranial stereotactic radiation therapy of small lung lesions using accurate dose calculation algorithms. Radiat Oncol 2006, 1: 45. 10.1186/1748-717X-1-45PubMed CentralView ArticlePubMedGoogle Scholar
- Fotina I, Winkler P, Künzler T, Reiterer J, Simmat I, Georg D: Advanced kernel methods vs. Monte Carlo-based dose calculation for high energy photon beams. Radiother Oncol 2009,93(3):645-53. Epub 2009 Nov 18 10.1016/j.radonc.2009.10.013View ArticlePubMedGoogle Scholar
- Koelbl O, Krieger T, Haedinger U, Sauer O, Flentje M: Influence of Calculation Algorithm on Dose Distribution in Irradiation of Non-Small Cell Lung Cancer (NSCLC) Collapsed Cone Versus Pencil Beam. Strahlenther Onkol 2004, 180: 783-788. 10.1007/s00066-004-1268-4View ArticlePubMedGoogle Scholar
- Fragoso M, Wen N, Kumar S, Liu D, Ryu S, Movsas B, Munther A, Chetty IJ: Dosimetric verification and clinical evaluation of a new commercially available Monte Carlo-based dose algorithm for application in stereotactic body radiation therapy (SBRT) treatment planning. Phys Med Biol 2010,55(16):4445-64. Epub 2010 10.1088/0031-9155/55/16/S02View ArticlePubMedGoogle Scholar
- Wulf J, Hädinger U, Oppitz U, Olshausen B, Flentje M: Stereotactic radiotherapy of extracranial targets: CT-simulation and accuracy of treatment in the stereotactic body frame. Radiother Oncol 2000,57(2):225-36. 10.1016/S0167-8140(00)00226-7View ArticlePubMedGoogle Scholar
- Takeda A, Kunieda E, Shigematsu N, Hossain DM, Kawase T, Ohashi T, Fukada J, Kawaguchi O, Uematsu M, Takeda T, Takemasa K, Takahashi T, Kubo A: Small lung tumors: long-scan-time CT for planning of hypofractionated stereotactic radiation therapy--initial findings. Radiology 2005,237(1):295-300. Epub 2005 Aug 18 10.1148/radiol.2371032102View ArticlePubMedGoogle Scholar
- Ahnesjö A: Collapsed cone convolution of radiant energy for photon dose calculation in heterogeneous media. Med Phys 1989, 16: 577-92. 10.1118/1.596360View ArticlePubMedGoogle Scholar
- Yamashita H, Nakagawa K, Nakamura N, Koyanagi H, Tago M, Igaki H, Shiraishi K, Sasano N, Ohtomo K: Exceptionally high incidence of symptomatic grade2–5radiation pneumonitis after stereotactic radiation therapy for lung tumors. Radiat Oncol 2007, 2: 21. 10.1186/1748-717X-2-21PubMed CentralView ArticlePubMedGoogle Scholar
- Mackie TR, Bielajew AF, Rogers DW, Battista JJ: Generation of photon energy deposition kernels using the EGS Monte Carlo code. Phys Med Biol 1988,33(1):1-20. 10.1088/0031-9155/33/1/001View ArticlePubMedGoogle Scholar
- Miften MM, Beavis AW, Marks LB: Influence of dose calculation model on treatment plan evaluation in conformal radiotherapy: a three-case study. Med Dosim 2002,27(1):51-7. 10.1016/S0958-3947(02)00088-2View ArticlePubMedGoogle Scholar
- Kawrakow I, Fippel M, Friedrich K: 3D electron dose calculation using a Voxel based Monte Carlo algorithm (VMC). Med Phys 1996,23(4):445-57. 10.1118/1.597673View ArticlePubMedGoogle Scholar
- Fippel M, Laub W, Huber B, Nüsslin F: Experimental investigation of a fast Monte Carlo photon beam dose calculation algorithm. Phys Med Biol 1999,44(12):3039-54. 10.1088/0031-9155/44/12/313View ArticlePubMedGoogle Scholar
- Fotina I, Kragl G, Kroupa B, Trausmuth R, Georg D: Clinical comparison of dose calculation using the enhanced collapsed cone algorithm vs. a new Monte Carlo algorithm. Strahlenther Onkol 2011,187(7):433-41. Epub 2011 Jun 27 10.1007/s00066-011-2215-9View ArticlePubMedGoogle Scholar
- Fippel M, Nüsslin F: Foundations of the Monte Carlo method for dose calculation in radiotherapy. Z Med Phys 2001,11(2):73-82.View ArticlePubMedGoogle Scholar
- Fogliata A, Vanetti E, Albers D, Brink C, Clivio A, Knöös T, Nicolini G, Cozzi L: On the dosimetric behaviour of photon dose calculation algorithms in the presence of simple geometric heterogeneities: comparison with Monte Carlo calculations. Phys Med Biol 2007,52(5):1363-85. Epub 2007 Feb 8 10.1088/0031-9155/52/5/011View ArticlePubMedGoogle Scholar
- Panettieri V, Wennberg B, Gagliardi G, Duch MA, Ginjaume M, Lax I: SBRT of lung tumours: Monte Carlo simulation with PENELOPE of dose distributions including respiratory motion and comparison with different treatment planning systems. Phys Med Biol 2007,52(14):4265-81. 10.1088/0031-9155/52/14/016View ArticlePubMedGoogle Scholar
- Chen H, Lohr F, Fritz P, Wenz F, Dobler B, Lorenz F, Mühlnickel W: Stereotactic, single-dose irradiation of lung tumors: a comparison of absolute dose and dose distribution between pencil beam and Monte Carlo algorithms based on actual patient CT scans. Int J Radiat Oncol Biol Phys 2010,78(3):955-63. Epub 2010 Feb 18 10.1016/j.ijrobp.2009.08.012View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.